Heart-type Fatty Acid-Binding Protein (H-FABP)Assay Kit
Accuracy: The quality control product is used for measurement. The average value of the measurement results should be within the range of quality control target values.
Precision: a, repeatability: The coefficient of variation(CV %) of the measured value should not be greater than 10 % with the same quality control or serum sample repeated 10 times.
B, interbatch difference: The interbatch difference for randomly selecting three batch kits should ≤ 10 %.
Linear: in[ 0.5, 130] In the ng/mL range, the linear correlation coefficient R ≥ 0.990, in[ 0.5, 50] The linear deviation within Ng / mL should not exceed ± 5 Ng / mL in(50,130) The determination deviation in the ng/mL range should not exceed ± 10 %.
Sensitivity: When the sample concentration is 20 ng/mL, the absorbance varies from 0.0100 to 0.1000.