Superoxide Dismutase（SOD）Assay Kit
Accuracy: The quality control product is used for measurement. The average value of the measurement results should be within the range of quality control target values.
Precision: a, repeatability: repeated determination of the same serum sample or quality control at different concentrations 10 times, and the coefficient of variation(CV %) of the measured value should not be greater than 10 %.
B, interbatch difference: the relatively poor inter-batch(R) for randomly selecting three batch kits should ≤ 10 %.
C, calibration bottle precision density(0 concentration point does not participate in the detection): calibration bottle precision should ≤ 10 %.
D, calibrate the precision between bottles: calibrate the precision between bottles should ≤ 10 %.
E, quality control bottle precision density: quality control bottle precision should ≤ 10 %.
F, quality control bottle precision density: quality control bottle precision should ≤ 10 %.
Linear: in[ 15, 250] In the U/mL range, the linear correlation coefficient R ≥ 0.99, in[ 15,150] The linear deviation within the U/mL range should not exceed ± 15U/mL in(150,250) The determination deviation within the U / mL range shall not exceed ± 10 %.
Sensitivity: When the sample concentration is 150U/mL, the absorption rate changes ≥ 0.0010.